Voltammetric Sensing of Endocrine Disruptor (Bisphenol A) Based on a Reduced Graphene Oxide/Carbon Nanotube/Gold Nanoparticles Nanocomposite Modified Screen-Printed Electrode
Herein, we present a facile and green approach to fabricate a highly sensitive electrochemical sensor for the determination of BPA in aqueous solution by using reduced graphene oxide (RGO), carbon nanotubes (CNT), and gold nanopaticles (AuNPs) modified screen-printed electrode (SPE). Graphene oxide (GO)/CNT nanocomposite was first deposited onto SPE, then electrochemically reduced to RGO/CNT at room temperature. Subsequently, AuNPs were electrochemically deposited in situ on RGO/CNT modified SPE. The morphology of bare SPE, GO/CNT-modified SPE, and RGO/CNT/AuNPs-modified SPE was characterized by scanning electron microscopy (SEM).
Figures 2a, b, and c displays the SEM images of the working electrode part of bare SPE, GO/CNT-modified SPE, and RGO/CNT/AuNPs-modified SPE, respectively. The bare SPE, shows sheet-like structures with many small particles dispersed throughout (Figure 2a). These structures can be assigned as graphitic carbon powder, according to the composition of the electrode. Figure 2b demonstrates the CNT could improve the surface roughness and potentially bridge the graphene sheet and small particles of graphitic powder. After electrodeposition of AuNPs, the AuNPs are seen evenly distributed in the RGO/CNT network (Figure 2c). Based on particle size analysis of at least 100 particles in SEM images, the average size of AuNPs was 84.6 ± 23.6 nm. Figure 3a displays the cyclic voltammograms of ferri/ferrocyanide solution at bare, GO/CNT-, RGO/CNT-, and RGO/CNT/AuNPs-modified SPEs. The electrochemical response at the bare SPE (at 50 mV·s-1scan rate) displays a relatively slow electron transfer and an irreversible electrochemical process. When modified with GO/CNT, the peak separation increased along with a reduction in the redox peak current. After GO is reduced to RGO (i.e., became RGO/CNT modified SPE), as expected, the redox peak current raised with a much smaller peak separation, which implies that most of the oxygenated groups on GO were removed and electron transfer was improved. The electrochemical response was further significantly enhanced after the electrodeposition of AuNPs, which is attributable to the extraordinary catalytic ability, large surface-to-volume ratio, and excellent electrical conductivity of the AuNPs.
Several important parameters, such as accumulation potential and time of differential pulse voltammetry, and pH value of buffer solution were investigated and optimized. Under the optimized conditions, the oxidation peak current in differential pulse voltammograms was proportional to BPA concentration in the range of 1.45 to 20 and 20 to 1,490 nmol/L of BPA, with an ultralow detection limit of 800 pmol/L (at signal/noise=3).
Figure 1. Fabrication of RGO/CNT/AuNPs modified SPE
Figure 2. SEM images of (a) bare, (b) GO/CNT-modified, and (c) RGO/CNT/AuNPs-modified SPEs
Figure 3. (a) Cyclic voltammograms at bare and GO/CNT-, RGO/CNT-, and RGO/CNT/AuNPs-modified SPEs. (b) Peak currents for BPA detection in two linear ranges of 1.45 to 20 nM and 20 to 1,490 nM. Inset is an expanded view of measurements of 1.45 to 20 nM BPA.
1. Yin, H., et al., Electrochimica Acta, 2010. 55(3): p. 603-610.