In this work, hydrothermal synthesis method was used for the fabrication of WO3 nanoparticles. For the fabrication of the pH sensor, the nanoparticles were mixed with Nafion and chitosan before drop coating onto the GCE surface. Scanning electron microscopy (SEM), x-ray diffraction (XRD), transmission electron microscopy (TEM) was used to characterize the nanoparticles. Cyclic voltammetry (CV) and square wave voltammetry (SWV) were used to monitor the potential shift based on different pH in the buffer solution. The WO3/GCE pH sensor showed a sensitivity of 1.15 VpH-1cm-2 at 25 °C and a potential drift of less than 5% after three hours of continuous use. The sensor showed good linearity at a pH range of 3 -10 and could retain 95% of its initial sensitivity after 1 week of use. The good sensitivity and long term stability of this WO3 based sensor may allow the development of a low cost pH sensor for a wider range of applications.3
This work is financially supported by Science Foundation Ireland funded project “SweatSens” under the grant agreement No. 14/TIDA/2455 and the University Grant Commission (research support and publication division), Bangladesh.
1. D. Vernardou, H. Drosos, E. Spanakis, E. Koudoumas, C. Savvakis, N. Katsarakis, J. Mater. Chem., 2011, 21, 513.
2. M. Natan, T. E. Mallouk, M. S. Wrighton, J. Phys. Chem. 1987, 91, 648.
3. L. Devlin, M. Jamal and K. M. Razeeb, Anal. Meth. 2013, 5, 880.